Technique feature of product
YTJ Electric Plaster Cutting saw is developed for the second
generation products of our company; it is the newly-designed
medical equipment products totally different from the first
generation. Compared to the first generation of products, this new
product considers the life-span of new material (such as the
polyester) and medical use instruments. It is revolutionary (been
improved from 20HZ to more than 133 HZ in the technical design of
operation frequency. it is more swift to cut. Distance slice of the
new instrument has adopted the new material; each of its technical
indicators is higher than the first generation.
The main capability parameters are as follows:
The maximum unload Output rev : >10000R.P.M
Swing angle: ≥4.8º
Rated input voltage : 220Vac±11Vac
Rated power: <125W
Noise: <85dB (When being unloaded)
1,Insert the plug in the outlet,hold the gypsum tightly to saw all
alone. Notice that the frequency of gypsum saw is more than 133hz,
so before using,don't close the switch of gypsum saw to avoid
2,Hold the gypsum with a firm grip and open the switch, if there is
no abnormity, put saw blade on the surface which need cutting, then
the vertical gypsum is cut down,if the gypsum material is the hard
polyester material, you can move the saw parallel to the gypsum
surface. If not sure, don’t cut entirely (although this gypsum has
the protect function for soft tissue.
3, vertically cut down the gypsum, lift the gypsum and move
forward, then cut the next section of gypsum saw surface.
4,Repeat the above-mentioned course,until finish the task.
5, If the cut section of saw is worn and torn, can unscrew the set
bolt with the special-purpose spanner,rotate an angle to tighten it
6, It is the same to the method of changing the saw blade as
7,If the frequency of output drops while using,such as the noise is
great,should generally stop continuing
8, Check the cable regularly while using to avoid using torn cable.
This Electric Plaster Cutting Saw is strictly forbidden using the
high temperature,high-pressure method to disinfect,can use soft
cotton cloth and neutral soap liquid to disinfect it. Forbid using
the organic solvent and rigid material. Don’t let the liquid into
the gypsum saw while cleaning.
If this Electric Plaster Cutting Saw can’t run, please check first
whether the carbon brush is worn and torn,in this way ,the ones
that should adopt our company’s special using carbon brush for
changing to prevent the armature from wearing and tearing
abnormally .If there are relevant problems, please contact us.
If there is abnormal failure in operating, it may lack lubricating
grease which can bear the high-temperature, can get in touch with
Should notice to dry it with hair-dryer after using, in order to
protect the electrical machinery, lengthen its span life .If there
are other problems, please getting in touch with us.