Coxa Joint Continuous Passive Motion Device , CPM Physiotherapy
Length : 60cm
Range of motion: -5-120 degree
Speed: 0.5-3.5 degree/s
Working time: 0-240min
Application: keen .coxa and ankle joint.
Motor power: 30 Q Mechanical Output
Maximum carrying force: 140kg.
Advantage of our factory :
1. High quality material, soft and strong and easily control
2. Customrized design and OEM are welcomed
3. 7 years' experience in export
4. More quantity ,more discount
5. Factory direct sale, competitive price
CPM Continuous Passive Motion for lower limb, Active cum passive
inbuilt with stimulator with Remote control
CPM Machine (CPM) is one of the primary methods for decreasing the
deleterious effects of immobilization and can deliver orthopedic,
neurological, and even circulatory benefits to the patient.
Immobilization, in turn, can create deleterious sequelae of
physiological and functional impairments.
The lower limbs CPM is used to help the patients to do
Rehabilitation suit for Keen, Coxa and ankle joint..
The ROM and working speed of the device is adjustable.
The knee joint of the device moves in constant angular velocity at
the adjusted value.
The patient can immediately interfere the working of the device
with the aid of pendant.
The device supports a large ROM with the aid of special design.
The device works safely and problem free with the aid of special
brush less dc motor.
Very powerful motor that can provide an 50 daN of horizontal force.
Reverses on excessive loads in order to protect patient safety.
Telescopic bed spacing bars in order to adjust the space between
the bed and device.
CPU control setting ,display and controlled by both machine and
Can automatically run by angle control or speed control
Can automatically controlled, including motion angle range and
Have the function of torque setting,
Use the motion support fitting for physiology curve
• The physician defines the protocol and ensures that it is
correctly implemented (adjustments, session time and frequency of
• We recommend that you lock the hand control when you give it to
• For optimum safety, always give the hand control to the patient
before starting the system.
• Check that the electrical socket is in good condition and is
suitable for the splint power supply cord.The latter complies with
current standards and has a grounding socket.The plug may be
connected to any standard socket.The socket must however have a
To connect the power supply,only use the original cable supplied
with the machine.
Check that the cables remain free around the device so that they do
not get damaged.